The liquid filter toluene in hexane (UV9) with reference filter toluene (UV9) shows in its spectrum in the range between 265 - 270 nm one distinctive feature, with which the spectral resolution resp. the actual slit width of a spectrophotometer can be determined according to the regulations of the European Pharmacopoeia and the U.S. Pharmacopeia.
Typical spectrum of the toluene liquid filter, measured with different slit widths
The spectral resolution is measured against the reference filter 667-UV9 hexane supplied with each filter set. The measurement is the same for all bandwidths. The wavelength range from 265 nm to 270 nm is scanned and the peak maximum and peak minimum are determined. Then the ratio between these two values is calculated.
|ARTICLE-NO.||Set: 667200 (UV6/UV9) Request this article |
Single filters: 667006 (UV6) / 667009 (UV9)
|APPLICATION||Checking spectral resolution according to EP and USP|
|CONTENT||Toluene in hexane (UV6) |
Reference filter hexane (UV9)
|STANDARD CERTIFICATION||Wavelength: Scan from 265 to 270 nm |
Slit widths: 0.5, 1.0, 1.5, 2.0, 3.0 nm
with Hellma Analytics calibration certificate (no DAkkS calibration certificate)
|POSSIBLE CERTIFICATION||Wavelengths: fix |
Slit widths: all from 0.5 nm to 3.0 nm