As is the case for all measuring devices, the glass filters used to verify spectrophotometers must also be checked and recertified at regular intervals.
In general, a recertification interval of 12 months is recommended for checking and recertifying glass filters during the first two years of use, with an interval of 24 months thereafter.
|666-S000||Set of glass filters: F0, F1, F2, F3, F4||666S000RE|
|666-S001||Set of glass filters: F3, F4, F7||666S001RE|
|666-S002||Set of glass filters: F2, F3, F4||666S002RE|
|666-S003||Set of glass filters: F1, F2, F3, F4, F7||666S003RE|
|666-S004||Set of glass filters: F201, F202, F203, F0||666S004RE|
|666-S005||Set of glass filters: F0, F1, F3, F4||666S005RE|
|666-S006||Set of glass filters: F0, F2, F3, F4||666S006RE|
|666-S300||Set of glass filters: F301, F303, F390||666S300RE|
Your filter set is not listed?
Request for the recertification of your filters directly from us.
Recertifications of UV / Vis reference materials from other manufacturers are also easily possible in our DAkkS calibration laboratory.